|Place of Origin:||CHINA|
|Minimum Order Quantity:||1set|
|Packaging Details:||export carton|
|Supply Ability:||200sets per month|
Enteral Feeding Pump is a transfer device which controlled by microcomputer .It can deliver gastrointestinal nourishment and common intravenous fluds in hospital and family.The advantage is an imitation of gastric motility rhythm. It is suitable for patients who without swallowing ability delivery nourishment to intestines or stomach.The pump with interrupt,required volume and many others alarm function, to ensure delivery safe and reliable.
Enteral feeding pump consists of shell,control panel,pump cover and peristaltic pump.Through peristaltic pump squeeze pump tubs to reach delivery nourishment.There is a microcomputer in the pump,it can control pump work status,flow rate,required volume and complete alarm,stop pump function.
Main Technique Parameters:
a) Infusion rate:0ml～999ml,≤±10%
b) Total volume:0.1ml～999ml,≤±10%
c) flow rate:1ml/h～300ml/h；
d) Infusion accuracy:At 1ml/h～50ml/h,range:1ml/h,≤±10%；
At 50ml/h～300ml/h,range 5ml/h,≤±10%
e) Power consumption:13VA
f) Size: 193mm×130mm×105mm
h) Feeding tube:Special feeding tube
Cumulative volume inquiry
Infuse rate setting
1) Daily cleaning:Using wet cloth dip a little detergent wipes the shell,then wipe off the detergent with a clean cloth.Finally, wipe with a dry cloth to dry.
2) Replace the fuse:Unscrew 4 screws in the back pump cover open pump,then unscrew the fuse casing to replace fuse.Fuse size:0.2A(Φ5×20).
3) When don't use,Please cleaning pump case and placed on dry shelf.
4) Please full charged for don't use for long time.And examination battery before next use,to ensure the battery can work.